Agilent Micro GC User Information
Embed Size (px)
Transcript of Agilent Micro GC User Information
Agilent 3000 Micro Gas Chromatograph
Agilent Technologies 2001, 2002 All Rights Reserved. Reproduction, adaptation, or translation without permission is prohibited, except as allowed under the copyright laws. Part number G2801-90118 First Edition, December 2002 Replaces part number G2801-90117, April 2002 and part number G2801-90110, July 2001. Made in USA Teflon is a registered trademark of E.I. du Pont de Nemours Inc. Swagelok is a registered trademark of Crawford Fitting Company. Stabilwax is a registered trademark of Restek Corporation. Software Licenses Portions of the software included with this product are covered by various free software licenses, including the GNU General Public License. Copies of these licenses are included in the \Licenses subdirectory on the enclosed CD-ROM. Software listed in the file public.txt is covered by one or more free software licenses. The specific license or licenses applying to a program or library is listed either at the head of the source modules for that program or library, or in a file named COPYING in the source distribution for that program or library. Software listed in the file agilent.txt is NOT covered by a free software license. Copyright notices for free software packages that require explicit acknowledgement are located in the \Copyrights subdirectory on the CD-ROM. For more information on licenses and copyrights applicable to the software used in this instrument, refer to file public.html in the root directory of the enclosed CD-ROM. Source code for the free portions of the included programs is available from Agilent for four years from the original date of purchase. Send a written request to the address below.
Safety Information The Agilent Technologies Micro Gas Chromatographs meet the following IEC (International Electro-technical Commission) classifications: Safety Class III, Transient Overvoltage Category II, Pollution Degree 2. This unit has been designed and tested in accordance with recognized safety standards and is designed for use indoors. If the instrument is used in a manner not specified by the manufacturer, the protection provided by the instrument may be impaired. Whenever the safety protection of the Micro Gas Chromatograph has been compromised, disconnect the unit from all power sources and secure the unit against unintended operation. Refer servicing to qualified service personnel. Substituting parts or performing any unauthorized modification to the instrument may result in a safety hazard. Hot Surfaces Should be Avoided The Micro Gas Chromatograph has heated inlets which are maintained at 110C. Contacting the inlets once they are at operating temperatures can result in injury. Extreme care should be taken to avoid these surfaces. Do Not Operate in an Explosive Atmosphere Do not operate the instrument in the presence of flammable gases or fumes. Operation of any electrical instrument in such an environment constitutes a definite safety hazard. Use Only Supplied Power Source Use of any other power supply could result in catastrophic failure of the electrical system leading to personal injury. Cleaning To clean the unit, disconnect the power and wipe down with a damp, lint-free cloth. Recycling the Product For recycling, send the product to: Agilent Technologies, Inc. 2850 Centerville Road Wilmington, DE 19808-1610 or Agilent Technologies Deutschland GmbH Hewlett-Packard Strasse 8 76337 Waldbronn Germany
Safety Symbols Warnings in the manual or on the instrument must be observed during all phases of operation, service, and repair of this instrument. Failure to comply with these precautions violates safety standards of design and the intended use of the instrument. Agilent Technologies assumes no liability for the customers failure to comply with these requirements. WARNING A warning calls attention to a condition or possible situation that could cause injury to the user. CAUTION A caution calls attention to a condition or possible situation that could damage or destroy the product or the users work. See accompanying instructions for more information Indicates a hot surface. Indicates hazardous voltages. Indicates earth (ground) terminal. Indicates explosion hazard.
Agilent Technologies, Inc. 2850 Centerville Road Wilmington, DE 19808-1610 USA
Electromagnetic compatibility This device complies with the requirements of CISPR 11 and EN 61326. Operation is subject to the following two conditions: 1 This device may not cause harmful interference. 2 This device must accept any interference received, including interference that may cause undesired operation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try one or more of the following measures: 1 Relocate the radio or television antenna. 2 Move the device away from the radio or television. 3 Plug the device into a different electrical outlet, so that the device and the radio or television are on separate electrical circuits. 4 Make sure that all peripheral devices are also certified. 5 Make sure that appropriate cables are used to connect the device to peripheral equipment. 6 Consult your equipment dealer, Agilent Technologies, or an experienced technician for assistance. 7 Changes or modifications not expressly approved by Agilent Technologies could void the users authority to operate the equipment. 8 This ISM device complies with Canadian ICES-001. Cet appareil ISM est conforme a la norme NMB001 du Canada.
Sound Emission Certification for Federal Republic of Germany Sound pressure Lp